Dec 31, Operating and service manual for the HPA Synthesizer/Function Generator. Complete with schematics. Who I HEWLETT mllnM PACKARD OPERATING AND SERVICE MANUAL MODEL A SYNTHESIZER/FUNCTION GENERATOR Serial Numbers: AH. HP / Hewlett Packard / Agilent Used A Synthesizer/Function Generator, Refurbished – Used Test Equipment at Test Equipment Depot.
|Published (Last):||24 June 2016|
|PDF File Size:||13.48 Mb|
|ePub File Size:||13.75 Mb|
|Price:||Free* [*Free Regsitration Required]|
Auxiliary Output Sine Function Only Actual display will depend greatly upon the accuracy of the oscilloscope amplifiers and display.
Change A frequency to kHz and function to square. Change A frequency to 1 MHz.
Each system can have only one system controller, even if several controllers have manuual control capability, Set A function to sine, frequency to 2 kHz, and amplitude to The following accessories are available for use with the Model A: Fewer parts and integrated circuit technology make the difference.
The sweep will stop while any other parameter is being changed, then wili restart.
Conversion to RMS or dBm is simple with the touch of a button. Used for a signal common, as well as providing protection against electrical shock in case of a fault.
Full text of “HP A Synthesizer/Function Generator Operating and Service Manual”
Maximum and minimum amplitudes are shown in Table Page 22 TB Figure 5. Go to subroutine to determine the reason. Electronic counter should indicate 10 When the instrument is turned on, the sweep mode is set to linear, and the parameters are set as follows: Connect A signal output to the AC voltmeter via a 6 ft.
Change attenuator to 12 dB. Change A function to square wave.
It is recommended that no other equipment be connected to this HP-IB during this test. Remove the ohm feedthru and connect the A output directly to the digital voltmeter 332a5. The controller requests device status from one device at a time. Frequency is always displayed in Hz, even though the entry may have been made in kHz or MHz. No circuit or layout changes.
HP A User Manual | 23 pages | Also for: B
Record frequency counter indication. The several cables can be connected to a single source.
Harmonic Distortion Verification Press the Negative Ramp function on the A. The A can be rack mounted in a rack having an El A standard width of A TTL compatible output is present during a sweep operation see Paragraph When in Mode 2, the EOS character should not be sent until the end of the program string or after 48 bytes; see Paragraph 3-H7.
3325A Synthesizer/Function Generator
The calculator prints the test results. Connect the A signal output to an oscilloscope -hp- A with a load. Programming Data Transfer Mode Verify that all three voltmeter readings are between 3.