Nov 24, Download a PDF version of the Axon Digidata Digitizer User Manuals. Axon Digidata B Axon Digidata A Digitizer User’s Guide. Nov 9, When I try to install the Digidata A under WinXP, I seem to be stuck in a loop . The Found New Hardware Wizard keeps appearing, and the. Nov 9, I think the wrong SCSI driver is loaded for the Digidata A. I cannot configure the digitizer, and it does not appear in the Windows Device.
|Published (Last):||24 December 2010|
|PDF File Size:||12.76 Mb|
|ePub File Size:||19.5 Mb|
|Price:||Free* [*Free Regsitration Required]|
By pressing ‘print’ button you will print only current page. Number of input channels: To print the manual completely, please, download it. The Power-on Holding Levels section is used to set the voltage on both Analog Out channels at the time that the Digidata A is turned on. If you purchased your instrument from a Distributor or OEM Supplier, contact them for repair service. New features also include better instrument telegraph support and line-frequency triggering. Trigger Output This is a pin female-type connector with the following pin assignments: If you have a Digidata A or B system that you still wish to use, it is OK to leave the old Digidata board in the computer and to leave DriverLinx installed during the Digidata A installation.
Secure the Rack Mount Panel to the rack with four rack screws not included. Works, clean, very good; previously owned, used. I’l have to check out a few things first as we’re in the UK and I’ll get back to you very soon. Products returned to our factory for repair must be properly packaged with transportation charges prepaid and the shipment fully insured. On the rear panel there are two SCSI ports, which are identical.
Page of 32 Go. Declaration Of Conformity EN This site seems to be very slow.
Axon™ pCLAMP® and DigiData™: Operating system compatibility
Hi Sludge Yes, still interested. Thanks for the response. Comments to this Manuals Your Name.
It was used in a research lab shut down now and I came across it from a friend. Scsi Termination ON position. We recommend that the Digidata A be calibrated whenever there is a change in the environment, such as moving it to another room or installing it on another computer. Please let me know.
Your Trusted Source for Used Lab Equipment
Don’t show me this message again. Buyer is responsible for determining the suitability of this product for their application. I would suggest starting your digidxta new topic if you 1232a just started shopping for a digidata yourself.
The higher bit dynamic range and lower noise makes a programmable gain unnecessary. TTL compatible Digital Outputs: If so can we move forward to buy it? The Digidata A does not include a programmable digitizer gain. HC compatible Output current: If your problem is still not resolved, please go through the “Before You Call” checklist below prior to contacting us. Hi Sludge Apologies I’ve been locked out and I’ve had to create a new account.
If it is determined your instrument must return to the factory for repair, the Technical Support Representative will issue a Return Merchandise Authorization RMA number. Is the A still for sale? Yes, I am still interested. It will work with pClamp 9.
Instruments if the problem persists. If you can reproduce a problem by following a series of steps using software, record these steps so that we can reproduce the problem. There are two LEDs: Please let me know and I will talk immediately to him and get back to you. Log in or register to post comments.
Secure the entire unit in the rack a. Chapter 2 Installation If you have a Digidata A or 132a system that you still wish to use, it is OK to leave the old Digidata board in the computer and to leave DriverLinx installed during the Digidata A installation. Copyright Axon Instruments, Inc. Page 10 Tighten the Side Bracket set screws up against the side panels. Enter text from picture: Power Supply The power supply is universal input.